Ag/Ag+ Reference Electrodes
Fast and secure
Non-aqueous Ag/Ag+ reference electrode
High quality reference electrode suitable for cyclic voltammetry electrochemistry cells
Non-aqueous Ag/Ag+ reference electrodes and aqueous Ag/AgCl reference electrodes are two of the most commonly used types for cyclic voltammetry and other electrochemical studies. Our low price non-aqueous Ag/Ag+ reference electrode makes no compromises on quality and are sold sealed in a ceramic frit junction.
The non-aqueous Ag/Ag+ reference electrode features a silver wire used for direct contact with 0.01M silver nitrate of acetonitrile solution (not included) sealed with a ceramic frit. Ag/Ag+ reference electrode is ideal when organic solvent such as acetonitrile is used in the electrolyte system.
On top of our site-wide large order discount offer, you can now save up to 5% per electrode when you buy a pack of five and up to 10% per electrode when you buy a pack of ten. A non-aqueous Ag/Ag+ reference electrode is also included with the Ossila Potentiostat and available in our amazing value electrochemical cell kits. Qualifying orders ship free as part of the Ossila Guarantee.
Ag/Ag+ Reference Electrode
High quality non-aqueous Ag/Ag+ reference electrode
Reference Electrode Multipacks
Buy reference electrode in pack of 5 to save 5% or pack of 10 to save 10%
Potentiostat with Cell and Electrodes
- Easy to Use
- High Accuracy + Precision
Worldwide Shipping £2100
All dimensions are in millimetres.
How to Prepare the Electrode
The reference electrode has a white cap which holds the silver or silver/silver chloride wire. The cap is detachable with a gentle force. Hold the glass tube and twist the white cap gently and pull it away from the glass tube. The glass tube is delicate so do not apply a strong force while holding the glass tube.
Once the cap is removed, use a syringe to inject the electrolyte to about 2/3 of the glass tube volume. We suggest gently flicking your index finger on the glass tube to get rid of any air bubbles in the electrolyte.
Put the white cap back on and rinse the reference electrode with acetonitrile. Wipe the electrode clean, air-dry for a few minutes, and now the electrode is ready to work.
Pricing and Options
|Non-aqueous Ag/Ag+ Reference Electrode
|Non-aqueous Ag/Ag+ Reference Electrode - Pack of 5
|Non-aqueous Ag/Ag+ Reference Electrode - Pack of 10
|Electrochemical Cell Kit with Electrodes
|Luggin Capillary Electrochemical Cell Kit with Electrodes
Electrode Polishing Kit EPK-3
- Restore Electrodes
- High Quality Polishing
- Great Value
Worldwide Shipping £165
To the best of our knowledge the information provided here is accurate. The values provided are typical at the time of manufacture and may vary over time and from batch to batch. Products may have minor cosmetic differences (e.g. to the branding) compared to the photos on our website. All products are for laboratory and research and development use only.